Free Space Calibration Procedure
I am trying to do some free space calibration to extract material data from unknown arbitrary material. I am struggling with the calibration part as the s parameters i am getting is not good enough to get proper value. Can anybody please suggest the calibration procedure?
I have E8361 keysight VNA. I have Type N short, open, load, through (50ohm) but no standard precise calkit. What should I do?
I would recommend that you perform a free space TRL calibration. The through is simply the distance between the two ports (antennas) without any sample. Reflect is a reflective plate in the sample holder. Line is the additional distance between the two ports depending on the measurement frequency - take the quarter wavelength of the mid-band frequency and move the second antenna by the corresponding length - example X-band - 9 GHz - 8 mm = line is through length + 8 mm.
Make sure you specify a sexless connector with a characteristic impedance of 377 Ohms.
Hope this helps.
Sign in or register to add a comment.
Add a Comment
About Our Site
Biological Effects and Applications
Computer Aided Design
Emerging Applications and Technology
Filters and Passives
MMIC and RFIC
Packaging and Materials
Sources and Receivers
Test and Measurement
Short and Open Standard in Calibration Kit
Impedance of free space
Complex characteristic impedance
Characteristic Impedance of Signal conductor
Terms of Service
Useful Hints and Tips
Created with PlushForums
© 2020 Microwaves 101 Discussion Board